|View single post by Bestgear|
|Posted: Sun Feb 19th, 2012 08:31 pm||
I hope someone can help and point me in the right direction.
Since having the gas analyser calibrated, the MTM (extended) on our BEA850 is no longer reponding.
I really think the guy doing the calibration does not have a clue as he went on about dry joints on the usb connector, which is crap, as the MTM has never been disturbed and always worked until he played with it.
Anyway - looking at the interfaces configuration page, the MTM interface was set to a non-existant com port!
The part number of the unit is 1 687 022 951 and from a search around, it appears to be an MTM-Extended, which is the device type selected, but when you unplug it from the PC and connect it again, the only device that gets added is a HID device - certainly not a COM port.
If anyone knows what MTM interface to select for this, or has a BEA850 and could have a look, I would really appreciate it, as no COM port is applicable being a HID device.
Thanks in advance.